+44 (0)1509 263171
Please use this identifier to cite or link to this item:
|Title: ||Assessment of the feasibility of the use of conductive polymers in the fabrication of ion mobility spectrometers|
|Authors: ||Koimtzis, Theodoros|
Goddard, Nick J.
Wilson, Ian D.
Thomas, C.L. Paul
|Issue Date: ||2011|
|Publisher: ||© American Chemical Society|
|Citation: ||KOIMTZIS, T. ... et al., 2011. Assessment of the feasibility of the use of conductive polymers in the fabrication of ion mobility spectrometers. Analytical Chemistry, 83 (7), pp. 2613 - 2621.|
|Abstract: ||The development of an ion mobility spectrometer with an injection molded plastic drift tube made from carbon-loaded nylon and the cyclo-olefinpolymer Zeonex is described. Thermogravimetric assessment combined with headspace analysis by ion mobility spectrometry and gas chromatography−mass spectrometry indicated that Zeonex encapsulated carbon-loaded nylon could be used to fabricate a snap-together injection molded stacked ring drift tube, 4.25 cm long that could be substituted for a conventional wire-wound heated ceramic drift tube of the same length into a high temperature ion mobility spectrometer. Temperature stability experiments indicated that such a combination of polymers produced stable water-based reactant ion peaks [(H2O)nH]+ up to a temperature of approximately 50 °C. Above this temperature, ammonia appeared to outgas, resulting in the production of [(H2O)n(NH4)mH]+ type species before, at higher temperatures, the release of oligomeric entities suppressed resolved ion responses. Surface charging effects were also observed, and over a period of continuous operation of 4 h, these caused suppression of the signal intensity (1.11−0.954 V) and an apparent mobility shift in the observed responses (K0 = 1.86−1.90 cm2 V−1 s−1). Substituting nylon, a polymer with a significantly lower surface resistivity, for the Zeonex demonstrated how surface charging phenomena could be managed though control of surface resistivity in future polymer formulations. The device was challenged successfully with test atmospheres of hexan-1-ol (K0 = 1.66 cm2 V−1 s−1 (monomer) and 1.32 cm2 V−1 s−1(dimer)) and dimethylmethyl phosphonate (K0 = 1.70 cm2 V−1 s−1 (monomer) and 1.44 cm2 V−1 s−1 (dimer)). The potential advantages of developing polymeric systems using more advanced polymer formulations are discussed.|
|Description: ||This document is the Accepted Manuscript version of a Published Work that
appeared in final form in Analytical Chemistry, copyright © American Chemical Society
after peer review and technical editing by the publisher. To access the final edited
and published work see: http://pubs.acs.org/doi/abs/10.1021/ac102997m|
|Sponsor: ||The work described in this paper was supported by a grant from AstraZeneca and took place as part of the National Initiative on Ion Mobility Spectrometry run at the former Department of Instrument and Analytical Science at UMIST and Nottingham Trent University with support from Waters Instruments, GlaxoSmithKline and AstraZeneca.|
|Version: ||Accepted for publication|
|Publisher Link: ||http://dx.doi.org/10.1021/ac102997m|
|Appears in Collections:||Published Articles (Chemistry)|
Files associated with this item:
Items in DSpace are protected by copyright, with all rights reserved, unless otherwise indicated.