+44 (0)1509 263171
Please use this identifier to cite or link to this item:
|Title: ||Compressed sensing current mapping methods for PV characterisation|
|Authors: ||Koutsourakis, George|
Hall, Simon R.G.
Betts, Thomas R.
|Issue Date: ||2016|
|Publisher: ||© IEEE|
|Citation: ||KOUTSOURAKIS, G. ... et al, 2016. Compressed sensing current mapping methods for PV characterisation. Presented at: The 43RD IEEE Photovoltaic Specialists Conference, 5th-10th June 2016, Portland, Oregon.|
|Abstract: ||The Compressed Sensing (CS) sampling theory has been combined with the Light Beam Induced Current (LBIC) method, to produce an alternative current mapping technique for photovoltaic (PV) devices. Compressive sampling of photocurrent is experimentally implemented using a Digital Micro-mirror Device (DMD). The main advantage of this new method for current mapping is that measurement time can be significantly reduced compared to conventional LBIC measurement systems. This is achieved mainly by acquiring fewer measurements than a raster scan would need and by utilizing the fast response of the micro-mirror array. Two different experimental layouts are considered in this work. The first is a small area optical set-up based on a single wavelength laser source. The second layout utilizes a commercial Digital Light Processing (DLP) projector through which compressive sampling is applied. Experimental results with both experimental schemes demonstrate that current maps can be produced with less than 50% of the measurements a standard LBIC system would need. The ability to acquire current maps of individual cells in encapsulated modules is also highlighted. The advantages and drawbacks of the method are presented and its potential to significantly reduce measurement time of current mapping of PV cells and modules is indicated.|
|Description: ||Personal use of this material is permitted. Permission from IEEE must be obtained for all other uses, in any current or future media, including reprinting/republishing this material for advertising or promotional purposes, creating new collective works, for resale or redistribution to servers or lists, or reuse of any copyrighted component of this work in other works.|
|Sponsor: ||This work was funded through the European Metrology Research Programme (EMRP) Project ENG55 PhotoClass. The EMRP is jointly funded by the EMRP participating countries within EURAMET and the European Union. This work is co-funded by the UK National Measurement System.|
|Version: ||Accepted for publication|
|Publisher Link: ||https://doi.org/10.1109/PVSC.2016.7749827|
|Appears in Collections:||Conference Papers and Contributions (Mechanical, Electrical and Manufacturing Engineering)|
Files associated with this item:
Items in DSpace are protected by copyright, with all rights reserved, unless otherwise indicated.